Measurement of radiation dose in multi-slice computed tomography

The aim of this study was to measure the radiation doses for computed tomography (CT) examinations of the head, chest and abdomen in adult patients in Nepal in comparison to international standard. Dose length products (DLP) and effective doses for standard patient sizes were calculated from the reported volume CT dose index (CTDI Vol ). Details were obtained from approximately 90 CT examinations carried out in 128 slice CT scan. Effective dose was calculated for each examination using CT dose indices, exposure related parameters and CTDI-to-effective dose conversion factors. The CTDI and DLP were below the established international reference dose levels for head and chest while for the abdomen and pelvis, the CTDl and DLP were above the established international reference dose levels. The mean effective doses in this study for the head, chest, and abdomen were 1.7, 5.4 and 17.7 mGy respectively. In conclusion, for the routine head and chest protocol, CTDI, DLP and ED were found to be significantly lower compared to the recommendation of European Commission. However, abdomen CT scans showed higher dose values because of multiple phase scans and longer scan lengths.


Introduction
Computed tomography (CT) is an imaging modality that produces cross sectional images representing the X-ray attenuation properties of the anatomical structures. 1 European Union, in an ionizing radiation protection directive, has classified CT as a high dose diagnostic procedure and has pointed to the need to reduce the dose to the patient. 3forts towards dose reduction in CT have been recommended by the International Commission on Radiological Protection (ICRP).The ICRP, the International Atomic Energy Agency, and the European Commission have all recommended the setup and the implementation of CT dose guidance levels for the most common CT examinations to promote strategies for the optimization of radiation doses. 4Computed Tomography Dose Index (CTDI) is the most widely used CT dose quantity, which integrates the long axis dose profile resulting from a single slice rotation of the X-ray tube.Another important dose quantity is the dose length product (DLP), which includes the patient, or the phantom volume irradiated during a complex examination.
The importance of radiation dose from X-ray CT has been underlined recently by the attention given in the scientific journals and literatures to issues of doses and the associated estimated risk.The dose levels contributed by CT exceeds those from conventional radiography and fluoroscopy and the usage of CT continues to expand, often by 10-15% per year. 5hus, CT will continue to contribute a substantial portion of the total collective dose delivered to the general public from medical examinations involving ionizing radiation.The rapid development of CT technology and the resultant explosion in new clinical applications, including cardiac CT, perfusion CT have created obligating need to teach, understand, and use CT dose information in more practical aspects.
The objective of this study was to measure CTDI and DLP during the most frequent CT examination.The mean values of computed tomography dose index weighted (CTDIw), DLP and effective dose were calculated for each protocol and were compared with the recommendations of European Commission.

Materials and Methods
This retrospective study was carried out from July to September 2015.CT performance information and examination parameters (for head, chest, abdominal and pelvic protocols) from the Teaching Hospital were collected.

Abstract
The aim of this study was to measure the radiation doses for computed tomography (CT) examinations of the head, chest and abdomen in adult patients in Nepal in comparison to international standard.Dose length products (DLP) and effective doses for standard patient sizes were calculated from the reported volume CT dose index (CTDIVol).Details were obtained from approximately 90 CT examinations carried out in 128 slice CT scan.Effective dose was calculated for each examination using CT dose indices, exposure related parameters and CTDI-to-effective dose conversion factors.The CTDI and DLP were below the established international reference dose levels for head and chest while for the abdomen and pelvis, the CTDl and DLP were above the established international reference dose levels.The mean effective doses in this study for the head, chest, and abdomen were 1.7, 5.4 and 17.7 mGy respectively.In conclusion, for the routine head and chest protocol, CTDI, DLP and effective dose were found to be significantly lower compared to the recommendation of European Commission.However, abdomen CT scans showed higher dose values because of multiple phase scans and longer scan lengths.For each protocol, 30 cases were collected.The patients that were included in the survey were selected in order to correspond to the typical patient (weight 40-70 kg).Examination other than head, chest and abdomen were excluded.Patients with weight out of range were rejected.Patients with gross abnormalities were rejected.

Results
CT scans of head (n=30), chest (n=30) and abdomen (n=30) were done.The doses to the patient were expressed in terms of CTDIvol, DLP and effective dose.

For head examination
The mean age of the patients (male 15, female 15) was 27.9 ± 8.2 (SD) years.The average weight and height of the patients was 56.4 ± 6.7 kg and 1.6 m respectively.The mean body mass index (BMI) was 21.8 ± 2.0 kg/m 2 .
The mean CTDIVOL before and after scan for routine head protocol was 45.9 ± 4.1 mGy and 45.5 ± 4.1 mGy respectively.The DLP before scan had a mean value of 756.7 ± 76.3 mGy-cm.The DLP after scan had an average value of 750.9 ± 77.3 mGy-cm.The mean effective dose was 1.7 ± 0.2 mSv.

For chest examination
The mean age of the patients (male 18, female 12) was 30.3 ± 13.7 years.The average weight of the patients was 56.4 ± 7.8 kg.The average height was 1.6 ± 0.1 m.The mean BMI was 21.6 ± 2.2.4 kg/m 2 .
The mean CTDIVOL before scan for non-contrast CT chest was 6.1 ± 1.9 mGy, and after scan was 5.6 ± 1.6 mGy (Table I).The mean CTDIVOL before scan for contrast enhanced were 6.0 ± 1.9 mGy and for after scan were 5.6 ± 1.8 mGy.The DLP before scan for non-contrast studies had a mean value of 215.0 ± 57.5 mGy-cm and for after scan were 189.6 ± 52.4 mGy-cm.The DLP before scan for contrast enhanced examinations had an average value of 219.6 ± 62.7 mGy-cm and for after scan, the mean value was 198.0 ± 60.2 mGy-cm.The average CTDI for both non-contrast and contrast enhanced studies was 11.2 ± 3.6.The mean DLP for both examinations was 390.5 ± 115.3.The mean effective dose was 5.4 ± 2.4 mSv.

For abdomen examination
The mean age of total 30 patients (21 male and 9 female) was 40.

Discussion
The CTDIW, DLP and effective dose were calculated in the patient who did not have any gross pathology.For head protocol, the mean CTDIVOL after scan, DLP after scan and effective dose were 45.5 ± 4.1 mGy and 750.9 ± 77.  6 The report of a CT survey indicated the mean DLP values for adult patients were ranged from 272-460 mGy-cm (head) 195-995 mGy-cm (chest), 270-459 mGy-cm (abdomen).Effective dose was calculated for each examination using CT dose indices, exposure related parameters and CTDI-toeffective dose conversion factors.CT air kerma index (CTDl) and dose length products (DLP) were below the established international reference dose levels.The mean effective doses for the head, chest, and abdomen were 0.82, 3.7 and 5.4 mSv respectively.Those values were observed that the effective dose per examination was lower in Sudan than in other countries.The difference was seen identified in CTDI, DLP and ED.
Abdullah (2009) measured Computed Tomography Dose Index in air CTDI (air) in Malaysian hospitals in 426 adult and 26 pediatric CT examinations. 7ffective doses for examinations of routine head, routine chest and pelvis were within the same range with studies conducted for the European guidelines, the UK and Taiwan.For the routine abdomen examination, the effective dose were still within the range compared to the studies for European guidelines and Taiwan, but 55.1% higher than the value from the study conducted in the UK.It provided the third quartile values of effective doses for every CT examination collected so that they could be used as reference in establishing the dose reference level of CT examinations in Malaysia. 7However, in this study, the CTDI and DLP were significantly higher than in UK for head, chest and abdomen scans.
Øberg (2011) measured the effective dose and its application to the medical field.Their objective was to calculate the effective dose (ED). 8In this study, these data promises to provide the exposure doses

Conclusion
For routine head and chest protocol, CTDI, DLP and ED were found to be significantly lower compared with the recommendation of European Commission (EC).Even though the CTDI, DLP and effective dose of the head and chest examination has been lower than the EC, these values did not affect the diagnostic image quality.For the abdomen and pelvis protocol the CTDI comparing with the EC was also lower.But, the DLP and effective dose for this examination is extremely high because of the very large scan length and multiple phase scans.For the abdomen examination, where the CTDI were found to be lower but the DLP and effective dose was higher, a revision of the scanning parameters using lesser scan length and less number of phase scanning is required whenever possible in order to reduce the value of effective dose.These data may be used as Dose Reference Level (DRL) values in Nepal.
Details were obtained from 90 CT examinations carried out in 128 slices CT scan (Siemens, SOMATOM Definition AS).DLP and effective | Original | Article |

Sudil Prajapati and Om Biju Panta dose
(ED) for standard patient sizes were calculated from the reported volume CT dose index CTDIVOL.Effective dose was calculated for each examination using CT dose indices, exposure related parameters and CTDI-to-effective dose conversion factors.The findings were then categorized and analyzed using SPSS version 21 and Microsoft Excel 13.The examination of head, chest and abdomen were only included.
Cite this ari le:Maharja S, Prajapai S, Pa ta OB.Measure e t of radiaio dose i ulisli e o puted to ography.Ba gaa dhu Sheikh Muji Med U i J. ; :-.Availa le at:. a glajol.ifoAJour al of Ba ga a dhu Sheikh Muji Medi al U i ersity, Dhaka, Ba gladeshMeasurement of radiation dose in multi-slice computed tomographySurendra Maharjan,